Laboratory and Industrial-Grade Thermometers
|Description of Services||Fee ($)|
|31010C||Organic Liquid in Glass Thermometers (0 ºC to 200 ºC) (32 ºF to 392 ºF)||271/pt|
|31040C||Organic Liquid in Glass Thermometers (-1 ºC to -110 ºC) (31 ºF to -166 ºF)||387/pt|
|31050C||Organic Liquid in Glass Thermometers (Liquid N2) (-196 ºC or -321 ºF)||271/pt|
|31100C||Quantity Tests of Liquid-In-Glass Thermometers||At Cost|
|31110C||Industrial Platinum Resistance Thermometers, Thermistor Thermometers, Digital Thermometers and Other Types of Thermometers (0 ºC to 150 ºC) (32 ºF to 300 ºF )||155/pt|
|31120C||Industrial Platinum Resistance Thermometers, Thermistor Thermometers, Digital Thermometers and Other Types of Thermometers (151 ºC to 315 ºC) (301 ºF to 600 ºF )||155/pt|
|31130C||Industrial Platinum Resistance Thermometers, Thermistor Thermometers, Digital Thermometers and Other Types of Thermometers (316 ºC to 550 ºC) (601 ºF to 1022 ºF )||155/pt|
|31140C||Industrial Platinum Resistance Thermometers, Thermistor Thermometers, Digital Thermometers and Other Types of Thermometers (-1 ºC to -110 ºC) (31 ºF to -166 ºF )||155/pt|
|31190S||Additional copy of Table from Results of 31110C-31150C at a Later Date||264|
|31200S||Preliminary Examination of Ineligible Thermometer||123|
|31260S||Special Thermometry Services, by Prearrangement||At Cost|
Fees are subject to change without notice.
This service provides for the calibration of a variety of thermometers covering the range from -196 °C to +200 °C (-321 °F to +392 °F). Thermometers belonging to the large and varied group which may be classified as laboratory, or "chemical," thermometers are regularly accepted. These are of the liquid-in-glass type with either solid-stem or enclosed scale. Ordinary household or meteorological thermometers will not, in general, be accepted unless the scale is graduated on the glass stem itself and the thermometer can be readily detached from its mounting for insertion in a liquid bath. Every thermometer submitted must be uniquely identified by a serial number and must pass a preliminary examination for fineness and uniformity of graduation; for cleanliness of the organic liquid and the capillary bore; for freedom from moisture, gas bubbles, and cracks in the glass; for adequacy or omission of gas filling where needed; for insufficient annealing; and, for misnumbered graduations. When these or other serious defects are found, the thermometer is returned untested.
The thermometers to be calibrated are placed in a constant temperature bath along with a NIST-calibrated standard platinum resistance thermometer (SPRT). The SPRT maintains calibrations traceable to the International Temperature Scale of 1990 (ITS-90), with a maximum expanded uncertainty of 0.7 mK.
Special tests may be conducted on temperature-measuring devices such as industrial grade platinum resistance thermometers, digital thermometers, and thermistors. Laboratory personnel should be consulted before submitting items.
ASTM Standard E 1, Standard Specification for ASTM Thermometers, Annual Book of ASTM Standards 14.03, p. 1, Amer. Soc. for Test. And Matls., Philadelphia, PA (1996).
ASTM Standard E 77, Standard Test Method for Inspection and Verification of Thermometers, Annual Book of ASTM Standards 14.03, p.61, Amer. Soc. for Test. and Matls., Philadelphia, PA (1996).
A Procedure for the Effective Recalibration of Liquid-in-Glass Thermometers, J. A. Wise, Natl. Inst. Stand. Technol. Spec. Publ. 819 (Aug. 1991).
Assessment of Uncertainties of Liquid-in-Glass Thermometer Calibrations at the National Institute of Standards and Technology, J. A. Wise, NISTIR 5341 (Jan. 1994).
The International Temperature Scale of 1990 (ITS-90), H. Preston-Thomas, Metrologia 27, 3-10 (1990); Metrologia 27, 107 (1990).
NIST Measurement Services: Liquid-in-Glass Thermometer Calibration Service, J. A. Wise, Natl. Inst. Stand. Technol. Spec. Publ. 250-23 (Sept.1988).
Thermometer Calibration: A Model for State Calibration Laboratories, J. A. Wise, and R. J. Soulen, Natl. Bur. Stand. (U.S.), Monogr. 174 (Jan.1986).Calibrations Phone: 301-975-5454, Fax: 301-975-2950 NIST,