Take a sneak peek at the new NIST.gov and let us know what you think!
(Please note: some content may not be complete on the beta site.).
NIST Authors in Bold
|Author(s):||David R. Leibrandt; James C. Bergquist; Till P. Rosenband;|
|Title:||Cavity-stabilized laser with acceleration sensitivity below 10-12/g-1|
|Published:||February 21, 2013|
|Abstract:||We characterize the frequency-sensitivity of a cavity-stabilized laser to inertial forces and temperature fluctuations, and perform real-time feed-forward to correct for these sources of noise. We measure the sensitivity of the cavity to linear accelerations, rotational accelerations, and rotational velocities by rotating it about three axes with accelerometers and gyroscopes positioned around the cavity. The worst-direction linear acceleration sensitivity of the cavity is 2(1) × 10-11/g measured over 0--50~Hz, which is reduced by 34~dB to below 10-12/g for low frequency accelerations by real-time feed-forward corrections of all of the aforementioned inertial forces. A similar idea is demonstrated in which laser frequency drift due to temperature fluctuations is reduced by 36~dB via real-time feed-forward using a temperature sensor located on the outer wall of the cavity vacuum chamber.|
|Citation:||Physical Review A|
|Pages:||pp. 023829-1 - 023829-7|
|Keywords:||Metrological applications,optical frequency synthesizers for precision spectroscopy,Resonators, cavities, amplifiers, arrays, and ring,robots,Servo and control equipment,Vibrations and mechanical waves|
|PDF version:||Click here to retrieve PDF version of paper (1MB)|