Take a sneak peek at the new NIST.gov and let us know what you think!
(Please note: some content may not be complete on the beta site.).
Calibration of Pitot Tubes in Turbulence
Today, the carbon dioxide emitted from the flue-gas stacks of coal-burning power plants is measured to control the emissions of the pollutants sulfur dioxide, nitrogen oxides (nitric oxide, nitrogen dioxide), and mercury. However, present-day carbon dioxide emission measurements may have uncertainties in the range 5% to 10%. NIST is conducting research to reduce the uncertainties to 1% or 2%. These lower uncertainties will be needed if the US Environmental Protection Agency (EPA) puts a price on carbon dioxide emissions, for example by a carbon tax or by cap-and-trade regulations.Carbon dioxide emissions are determined by mapping the flow in stacks using 1-, 2- and 3-dimensional differential pressure devices, such as standard Pitot static tubes, S-probes and multi-hole Pitot tubes. NIST has outfitted its low-turbulence wind tunnel with an automated stage to calibrate such differential pressure devices as a function of air-speed, pitch angle, and yaw angle. During preliminary measurements, Crowley, et al. discovered that the calibration can show hysteresis that gradually disappears as the turbulence in the flow increases. Thus calibrations must account for the turbulence in the flow to achieve the target accuracies of 1% or 2%.
In addition to improving the calibration of differential pressure devices, NIST is developing a scale-model smoke stack test-bed to study calibration protocols. NIST is also studying unconventional techniques for measuring carbon dioxide emissions such as a long-wavelength acoustic flow-meter.